the Cool.Click 2?
Cool.Click 2 is an HGH Injection Device designed to deliver Serostim
Brand Human Growth Hormone. Both Serostim and the Cool.Click 2 are
pharmaceutical products produced by EMD Serono.
Makes the Cool.Click 2 Unique?
Cool.Click 2 is one of the few Human Growth Hormone Injection Devices
which do not use needles, making it an ideal choice for both
pediatric and adult patients that wish to experience the benefits of
HGH Hormone Replacement Therapy without having to stick themselves
with a needle every single day.
Does the Serostim Cool.Click 2 Work?
Cool.Click 2 is able to deliver Serostim HGH without a needle because
it is capable of firing Liquid Serostim directly into the skin using
a pressure system. This Serostim Device launches a stream of
Bio-Identical Human Growth Hormone powerful enough to pierce directly
through the upper layers of the skin, into the subcutaneous layer
where all HGH Medications are delivered.
Cool.Click 2 is able to achieve all of this mechanically through the
combination of a high-powered spring which controls a piston which
forces the Serostim Growth Hormone to move at high velocity. In order
to prep for your daily HGH Dose, simply wind a trigger on the device
to coil up the spring, and press the trigger on the side of the
device as the tip is pressed against the skin to launch the Serostim
into your system!
are the Advantages of the Cool.Click 2
As we mentioned previously, the Cool.Click 2 is highly desirable
because it does not use needles to deliver the injections. The
product is also great because, once you learn how to use it, you can
quickly and easily go about your daily HGH Injections.
are the Disadvantages of the Cool.Click 2?
the Cool.Click 2 is a complex mechanical device with electronic
components, it is more expensive than Pens and other HGH Products. It
also does have an initial learning curve, but once patients
understand the process, it is very easy to use.
I Buy the Serono Cool.Click 2 for Needle-less Human Growth Hormone
The Cool.Click 2 is one of a multitude of Human Growth Hormone
Devices and Products that the Conscious Evolution Institute has to
offer. If you are interested in this HGH Injection Device, we can
have it shipped to your home, as well as the necessary HGH Cartridges
or HGH Vials, as soon as we diagnose your need for Bio-Identical
Human Growth Hormone Therapy.
We also offer a highly competitive pricing structure, so you can get
your HGH Device for the most efficient price. We also offer a wide
array of payment plans to make it easy to get the treatment you need
to restore your natural Hormone Balance.
Cool.Click 2 Device Instructions
Set Your Dosage with the Cool.Click 2
With one hand, hold the base of the device. With the other hand,
grip the front of the device and its cap. Turn the base of the
Cool.Click 2 until the two ends meet.
Take off the cap and turn on the device by pressing the On/Off
A key comes with the device, designed to help you set the dose. Push
the key into the corresponding keyhole on the front casing of the
device until the two prongs on the key activate the Dose Settings.
Remove the key and set aside.
Check the display. It should be flashing.
Briefly press the On/Off button. From here, you can press the button
as needed to change the dosage settings to those which have been
recommended for your HGH Replacement needs.
After you have chosen the setting that you desire, use the key
again. Upon activation, it will lock in your chosen settings, and it
will remember these settings no matter what until you go through the
same process again.
Attach the Piston and Nozzle to the Cool.Click 2
Open the package which seals the assembly for both the piston and
Take the Cool.Click 2 HGH Device and hold the front and back
portions of the device. Rotate the back end of the device so that
the front and back lock into place.
Take off the cap and draw back the locks which are designed to hold
the nozzle in place.
Push the piston and nozzle in place so that the piston will rest
against the spring.
Press down on the Cool.Click Nozzle until the piston locks into
Release the locks, so that they return to place, holding the nozzle
and piston in place. Make sure you hear a click.
Make sure you built the device correctly. Tug gently on the nozzle
and make sure it is one with the rest of the device. Don't take off
the nozzle until you are ready to replace it.
regarding the preparation of Serono HGH Cartridges and Vials
The Cool.Click 2 is designed to work with most Serono Recombinant HGH
Products, including Serostim Vials and Cartridges. All forms of HGH
Injection Therapy come with specific instructions pertaining to the
reconstitution of the lyophilized HGH solution.
The way that Serono HGH Products are connected to the Cool.Click 2 is
via connectors. There are two forms of connector, one of which is
designed to fit vials, the other of which is designed to fit
cartridges. When you receive the connector, it will be enclosed in a
sealed or sterilized package. If the package appears tampered with,
or is torn, do not use the product.
Before attaching the connector, make sure that the product is not
expired. After you make sure your product is in date, follow the
specific instructions outlined for the individual HGH Product.
Attach the Human Growth Hormone Vial or Cartridge to the Cool.Click 2
Hold the front and back of the Cool.Click 2, with the cap attached,
and twist the device so that both sides are connected.
Take off the cap and turn the Serostim Cool.Click 2 on.
Take off the protective shield from the vial or cartridge connector.
Link the specific connector with the nozzle on the device.
Handle the Cool.Click 2 facing up and down, with the vial or
cartridge facing the ceiling. Make sure you are looking directly at
the display. Turn the base of the HGH Device in a manner in which it
will start to take Serostim HGH into the nozzle. The screen will
show exactly how much Bio-Identical Human Growth Hormone that is in
After your dose is showing on the screen, wind the device one more
complete turn and gently agitate the nozzle to free any clinging
bubbles, then rewind the device to evacuate the bubbles. Repeat the
wind-unwind cycle until no more bubbles are present.
Remove the Vial or Cartridge from the Cool.Click 2 and recap the
Administer Your Serostim Dose with the Cool.Click 2
Screw the rear casing so that it links fully with the front casing
of the device. You may or may not hear a click as it screws in
Turn on the Cool.Click 2. Make sure the display reads 0.00. If you
see three eights flashing, this means that the device is not
properly tightened. Repeat step one.
Attach the nozzle of the HGH Device to your vial or cartridge
Wind the device until it draws the appropriate dose of Serostim HGH.
The dose will show up on the display.
Wind the Cool.Click 2 one extra time. Flick or tap at the side of
the device, so that any trapped air bubbles float up to the tip of
Rewind one rotation in order to return to your ideal dose, then
check the nozzle for air bubbles once again. If you still see
bubbles, return to step five and repeat until the nozzle no longer
Pull the nozzle away from the Cool.Click 2 Vial or Cartridge
Connector. Your dose is ready to be administered.
Clean the area where you intend to deliver the Serostim HGH
Needleless Injection with an alcohol swab. Be patient for the
wetness to dissipate.
Press the nozzle to your chosen administration site. Firmly click
the trigger at the top of the device, while pressing toward the skin
in order to release the Serostim HGH into your system.
Keep the nozzle pressed against the skin for around three seconds
before moving the Cool.Click 2 away.
Apply pressure to the HGH Injection Site with sterile gauze for
around thirty seconds.
Wipe the nozzle clean with a new alcohol swab. Cap the Cool.Click 2
and Return the Serostim Cartridge or Vial to the Refrigerator. Don't
refrigerate the Cool.Click 2. Keep it in a dry, safe place.
Take Care of Your Serostim Cool.Click 2
- The Cool.Click 2 should be stored in a safe environment between 50
and 104 degrees.
- Store the product in a relatively dry environment.
- Always keep the Cool.Click 2 Device clean. It can be washed with
mild detergent and a moist cloth.
- Never get the internal components of the Cool.Click 2 wet. Never dip
- Don't place the HGH Delivery Device in direct exposure to the sun.
- Do not refrigerate the Serono Cool.Click 2.
- Keep the device in its storage container when not using it.
- This Serono HGH Device is only designed to be used for three years.
Discard after three years and arrange to have a new device shipped
Cool Click 2 Helpful Tips
Before injecting Bio-Identical Serostim HGH, gather your tools in a
dry, clean area. The following are the tools that you'll need to
perform a routine injection:
- Alcohol Swabs
- Sterile Cotton or Gauze
- Serostim Reconstitution Kit or HGH Cartridge Connector
- Piston and Nozzle
- The Cool.Click 2
- Never activate the Cool.Click 2 while the nozzle still contains air.
- Never activate the Cool.Click 2 without first connecting the nozzle.
- Before every injection, check the dose in the nozzle to ensure that
no air is caught between the nozzle and the piston.
- If you accidentally put too much Serostim HGH in the nozzle, just
push the HGH back into the vial or cartridge and restart the
- Always leave the Cool.Click 2 capped when you aren't using it
- Always follow your physician's advice regarding HGH Replacement
- Always read all documentation with your medical treatment